Things I Discovered with My Son's CI Sound Processor
By Eric Sherman
My son uses an Advanced Bionics
We found over time water leaked into the battery compartment, and the battery would begin to rust. This was very concerning to us. We thought how could the seal start breaking down so quickly? What we discovered was the battery cap was cracked.
We found the cap had developed a micro crack on the end. You could only see it when you pulled outward on the sides of the cap (pictured). The crack most likely developed over time from daily use, and from attaching the processor clip.
We decided to replace the daily use caps with new caps dedicated just for swimming. We picked a different color not to mix up the caps. Ever since we started changing caps for swimming, we have not had any problems with water leaks.
If you find water getting into your battery compartment, check for small cracks and make sure the gasket around the processor is free of debris. Also, consider using different caps when swimming if you use the
processor for everyday use.
What have you discovered with your processors? Share your experiences; we’d love to hear them.
#CiWear is a shirt with pockets designed for #cochlear implant (CI) processors and other listening devices. Use as a rash guard, #swim shirt or an exercise shirt.